XLW（PC）Auto Tensile Tester
- General Details
♦ 0.5% of full scale can effectively ensures the test results accurately.
♦ adjustable measuring range, system integration, easy to change, the test can be resumed after replacement.
♦ Original mechanical vision and super-lenghth journey could meet the large deformation test.
♦ The instrument is controlled by micro-computer with menu interface, PVC operation panel and large LCD display
♦ 7 independent test modes are available, including tensile strength, peeling force, tearing force and other force properties of flexible packages
♦ 1000mm extra-long stroke satisfies the test needs of various materials with extreme elongation rate
♦ Different test range of load cells and 7distinct test speeds to meet different test requirements
♦ Intelligent designs of over-travel protection, overload protect, automatic position reset and power failure memory for safe test operation
♦ Professional operating software support statistical analysis of group specimens, superposition analysis of test curves and historical data comparison functions
- Principle & Standards
he pre-conditioned specimen is mounted between two grips, which move in relative direction during the test. The changes of force and displacement are separately recorded by the load cell fixed on the driven grip and embedded displacement transducer. The tensile strength, tear strength and elongation rate can be obtained by further calculation.
ISO 37、GB 8808、GB/T 1040.1-2006、GB/T 1040.2-2006、GB/T 1040.3-2006、GB/T 1040.4-2006、GB/T 1040.5-2008、GB/T 4850-2002、GB/T 12914-2008、GB/T 17200、 GB/T 16578.1-2008、 GB/T 7122、 GB/T 2790、GB/T 2791、GB/T 2792、GB/T 17590、ASTM E4、 ASTM D882、 ASTM D1938、ASTM D3330、ASTM F88、 ASTM F904、JIS P8113、QB/T 2358、QB/T 1130
- Applications Specifications
Basic Applications Extended Applications (Additional Accessories Required) Tensile Property Puncture Test of HypodermicNeedles in Artificial Skin Puncture Test of Films Puncture Test of Infusion Bags Puncture/Pullout Test of Flexible Rubber Closures Test of Tensile Strength and Elongation Rate Opening Resistance Test of Combined Covers Tear Test of ZD -Type Caps Opening Force Test of Oral Liquid Caps Puncture/Pullout Test of Oral Liquid Caps Test of Tensile Strength at Break 90 Degree Pullout Test of Infusion Bag Caps Pullout Test of Infusion Bag Caps Puncture/Pullout Test of Bottle Caps or Rubber Closures 23 Degree Pullout Test of Bottle Caps TearResistance Test 90 Degree Peel Test of Adhesive Tapes Tear Resistance Test of Adhesive Binding Books 90 Degree Peel Test of Water-soluble Plasters Tear Resistance Test of Adhesives Heat Seal Strength Test Adhesive Strength Test (soft) Adhesive Strength Test (hard) Peel Test of Flexible Tube Caps Removal Force of Pipes and Pipe Joints 90 Degree Peel Test Pullout Test of Cosmetic Brush Hair Pullout Test of Tooth Brush Hair Tensile Strength of Ropes at Break Opening Force Test of Jelly Cups and Yogurt Cups 180 Degree Peel Test Peel Test of Cup Films Pullout Test of Rubber Stoppers 45 Degree Peel Test of Bottle Membranes Tensile Strength of Sealing Bags Peel Test of Magnetic Cores Peel Test of Magnetic Cores Tear Resistance of Heat Sealing Films Separating Force of Protection Films 20 Degree Peel Test Peeling Grips of Floating Rollers Eccentric Grips Unwrapping Force of Adhesive Tapes Wide Sample Grips 135 Degree Peel Test of Plugs Tear Test Using Trouser Method Compressive Resistance of Plastic Bottles
- Technical Specifications
Specifications XLW (PC) Load Cell Capacity 500 N;50N (1000 N are optional) Accuracy 0.5% FS Test Speed 50, 100, 150, 200, 250, 300, 500 mm/min 1 Specimen Width 50 mm (Standard Grip) Stroke 1000 mm Instrument Dimension 450 mm (L) x 450 mm (W) x 1410 mm (H) Power Supply AC 220V 50Hz Net Weight 68 kg
- Related Products